Why am I getting noise in my readings? Print
Modified on: Tue, 13 May, 2014 at 3:10 PM
One of the primary causes of noise in readings is a BNC cable that exceeds recommended lengths. The Eon™ contains a built-in oscillator that will sustain clear readings up to one foot (30 cm) (See illustration below). Exceeding this length will cause inaccurate readings.
Note: An external oscillator, which Colnatec offers for purchase, will give you an additional 30 cm of length between crystal and controller.
Did you find it helpful? Yes NoSend feedback
Sorry we couldn't be helpful. Help us improve this article with your feedback.